General and Pricing questions :
What are the BioSemi Standard terms & conditions of sale ?
What technical support does BioSemi provide ?
How much does a BioSemi system cost ?
Software and interface questions :
How to install the USB interface and its drivers ?
How to install the USB drivers under Windows 8 or Windows 10 ?
How can I add additional features to ActiView ?
How can I make my own acquisition software, interfacing direct with the USB2 receiver ?
Which software is standard delivered with the system ?
Which file format does BioSemi use ?
Do we need LabVIEW to be able to run the BioSemi acquisition software ?
Can we integrate our existing LabVIEW analysis software with the BioSemi acquisition software ?
How are stimulator trigger signals handed by BioSemi hard and software ?
Does BioSemi produce an amplifier that we can use with our existing AD card ?
How can I review the ActiveTwo its perfect Trigger timing ?
Presentation or E-Prime cable to BioSemi receiver ?
USB Trigger Interface Cable
Active electrodes questions :
Can I use BioSemi active electrodes with my existing amplifier ?
Is skin preparation necessary when using BioSemi active electrodes ?
Can we use the BioSemi active electrodes without paste ?
What kind of input protection do the BioSemi active electrodes have ?
What are the advantages of Active electrodes with respect to shielding ?
Amplifier and AD-converter questions :
Is it possible to adjust the amplifier gain on BioSemi systems ?
How does BioSemi limit the current flow to the subject ?
Can we adjust the filter settings on a BioSemi system ?
Is the sample-rate adjustable on a BioSemi system ?
What are the advantages of a 24 bit system ?
Why is battery power-supply in combination with fiber data transfer so important ?
What is the function of the CMS and DRL electrodes, Ground, Reference or what ?
What is the smallest signal we can measure with the ActiveTwo bitstep of 31nVolt ?
What are the advantages of one ADC per channel ?
AD-box shuts down or acquisition stops
Practical use questions :
How to "split" the flat-cables ?
How to check an electrode set for defects ?
Battery and charger operation guidelines and replacement.
Pin-layout of the connectors on the AD-box
Pin-layout of the 37 pin Sub-D connectors on the AD-box with passive inputs.
Pin-layout of the connectors on the AIB
Pin-layout of the Trigger connector on the receiver.
Speedmode switch on the AD-box explanation.
Position coordinates of the electrodes for BioSemi headcaps.
How to make your headcap ?